PRECISE angle alignment of toric IOLs
During implantation of toric IOLs precise angle alignment of the lense in the capsule is required For this procedure TOCULAR is offered by HAAG-STREIT SURGICAL.
TOCULAR is a rotatable wide angle ocular with an integrated reticle that can be rotated manually via a sterilizable knob. For best visibility the reticle is clear in the center of the visual field. It allows an IOL alignment of better than +/- 2°.
Marking of the eye
Prior to the operation the astigmatic angle has to be determined with a keratoscope or a biometer. We recommend the use of the Haag-Streit LENSTAR LS 900. The vertical and/or horizontal coordinates are ink-marked on the eye with a sterilizable pre-marker as the Davis pre-marking system distributed by John Weiss.
Alignment of the toric IOL
After implantation of the IOL, the microscope zoom is adjusted to show the ink-marking in the visual field of the oculars. With the knob on the TOCULAR the reticle is turned to align the large scaled markings with the ink-marks on the eye. Now the IOL can be brought into position using the medium and small reticle markings.